Received Date: August 28, 2012; Accepted Date: October 16, 2012; Published Date: October 18, 2012
Citation: Okada J, Horiuchi H, Hashimoto K, Hirosawa D, Kurosaki Y, et al. (2012) Mobile Automatic Detection System for Bacillus anthracis using Electrochemical DNA Chip. J Biosens Bioelectron 3:126. doi: 10.4172/2155-6210.1000126
Copyright: © 2012 Okada J, et al. This is an open-access article distributed under the terms of the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original author and source are credited.
Visit for more related articles at Journal of Biosensors & Bioelectronics
A novel detection system for biological agents using Loop-Mediated Isothermal Amplification (LAMP) and an electrochemical DNA chip, Bio BulwarkTM, has been developed. Genomic DNAs/RNAs roughly extracted from biological agents were amplified by the LAMP method using target-specific primers and strand-displace polymerase. The amplified DNAs were processed to hybridization with the electrochemical DNA chip. Anodic current derived from hoechst 33258, an electrochemically active intercalator, was relative to the amount of the target DNA hybridized with target-specific probes on the DNA chip. All reactions and measurements were conducted using disposable cassettes, which included the reagents for amplification and the DNA chip, and a mobile automatic detection system. The preliminary study showed that accurate detection of B. anthracis was carried out within 70 minutes. This system would prevent or minimize damage in the event of a criminal attack involving a biological threat.
B. anthracis; Loop-mediated isothermal amplification; Mobile system; DNA chip cassette
Rapid identification of biological warfare agents on site is important for preventing or minimizing damage in the event of a criminal attack involving a biological threat. Although various detection systems have been developed, there are few useful systems available at present in terms of sensitivity and plurality of detection targets [1-3].
Molecular techniques based on an enzymatic amplification such as polymerase chain reaction are powerful tool for the sensitive and specific detection of microorganisms. Recently Loop-Mediated Isothermal Amplification (LAMP) has developed and become popular due to high specificity, high sensitivity and easy handling [4,5]. There have been a growing number of reports on the detection of dangerous bacteria or viruses using the LAMP method [6-11].
A DNA chip is a powerful tool for detecting multiple genes simultaneously. Although DNA chips based on a fluorescence detection method are already in widespread use [12-15], they require complicated procedures for fluorochrome labelling. Furthermore, expensive and large optical equipment is necessary. Recently electrochemical detection methods have been developed to overcome the above-mentioned drawbacks . It uses an electrochemically active intercalator as amplification or hybridization indicator and requires no labelling step. The electorochemical signal of intercalator is measured by micro electrode and small size potentiostat, therefore large and expensive equipment is unnecessary. A combination of LAMP and the electrochemical detection is reported to achieve a simple genetic testing [17-20].
Here we report a newly developed mobile system for the detection of biological agents, Bio BulwarkTM, which performs the entire process from DNA amplification to detection automatically using the LAMP method and an electrochemical DNA chip.
Environmental samples include impurities that inhibit DNA amplification and detection on the DNA chip. Furthermore, B. anthracis, a high-priority biological agent, exists as a spore in the field. The spore is a resistant form against unfavorable conditions such as high temperature, high pH, dry condition and irradiation. The extraction of DNA from the spore is difficult because it has a very hard shell. We developed a pretreatment reagent kit for environmental samples to extract and purify the DNAs, which is based on the beads beating method . The kit consists of a sampling tool (SWAB TEST Pro. media ST-25, Elmex Ltd.), zirconia-silica beads (φ0.1 mm, Tomy Medico Co., Ltd) for beads beating and isopropanol and ethanol for purification. First, powder, liquid or deposit samples are collected and suspended in a buffer solution using a sampling tool. The suspended samples are added to a bead beating tube to crush a cell. The tube is shaken vigorously using a mini bead beater (Biospec 3110BX ) for 3 min. After centrifugation at room temperature, a supernatant is moved to another tube and heated at 80°C for 3 min. Then, isopropanol and ethanol are added to the tube to precipitate and purify the DNA. We utilize B. subtilis as a sample in order to estimate extraction efficiency and purity of extracted DNA using the pretreatment kit, because it belongs to the same genus as B. anthracis and handling of the experiments is easier and safer. The extracted DNA was measured by a spectrophotometer.
The reaction condition of LAMP for B. anthracis was as described in a previous paper . The specific primers for three target genes are designed: one is a unique gene in the genomic DNA (gDNA) and the other two are virulence plasmids, pXO1 and pXO2. The isothermal amplification was performed at 63°C.
Figure 1 presents the principle of electrochemical DNA detection system . The electrochemical DNA chip consists of gold electrodes patterned on glass substrate and DNA probes immobilized on electrodes. By applying target DNA on the DNA chip, the DNA probe with the complimentary sequence specifically hybridizes with the target DNA. After washing off the non-specifically bound DNAs, electrochemically active intercalator Hoechst 33258 is applied on the chip. The intercalator specifically binds to the double-stranded DNAs, i.e. hybridized DNA probes. When voltage is applied to the electrode, the intercalator is oxidized and emits electrons. Anodic current derived from Hoechst 33258 was relative to the amount of the target DNA hybridized with target specific probes on the electrode. Thus, the target DNA is detected simply by measuring an electrical current. DNA probes for three target genes of B. anthracis immobilized on the electrode surface were designed for a loop sequence of each LAMP product in accordance with a previous paper . Oligonucleotide probes with a thiol group at 3’ end were obtained as custom synthesis products. Each working electrode was spotted with 0.1 μL of the probe solution containing the 3 μM oligonucleotide probes. Then, the chip was washed with distilled water and stored at -20°C. For each probe, four electrodes were assigned. The hybridization and washing reaction were controlled at 45 and 30°C, respectively.
DNA chip cassette
The DNA chip cassette has functions for both DNA amplification by LAMP reaction and DNA detection by electrochemical DNA chip. As shown in (Figure 2a), on the top surface of the cassette, there is a sample injection port and electrical signal connecting pads. This cassette has liquid flow channels and accordion chambers that serves as the reagent reservoir and the pump, respectively. After injecting a sample into a cassette, the cassette is completely sealed and all reactions are thereafter performed within the airtight cassette in order to prevent contamination by leakage of amplified DNA products. There are eight chambers in the cassette as shown in (Figure 2b). The first chamber is to hold the injected sample (sample chamber), the next four chambers contain DNA primers and polymerase for multiplex amplification (amplification chambers), and the other chambers contain reagents for hybridization, washing, and electrochemical detection (reagent chambers). Sample and reagents flow through narrow channels in the cassette, and microvalves in the channels control the flow direction. All valves and pumps are controlled from outside. Figure 3 shows the schematic diagram of the sample and reagent flows and the reaction process in the cassette. An injected sample is divided into 4 amplification chambers that are heated for DNA amplification. All amplified DNA products are gathered into a mixing chamber and mixed with hybridization reagent stored in the chamber in advance. The mixed sample is transferred to the DNA chip and hybridized with DNA probes on the DNA chip. Then, a washing buffer removes non-specified hybridized DNA. Finally, intercalator molecules react with double-stranded DNA on the DNA chip and electrochemical measurement is performed. Apart from injection of the pretreatment sample, all processes were performed in the cassette without preparation of reagents.
Mobile fully automated DNA detection equipment
Figure 2c shows the mobile fully automated DNA detection equipment Bio BulwarkTM for the DNA chip cassette described above. The entire system is incorporated in a compact and portable case. This equipment has a touch screen LCD for easy operation. A maximum of 15 kinds of biological agents are detectable simultaneously by using 2 cassette slots. Detection time is 70 min in high-sensitivity mode and 40 min in high-speed mode. All systems are controlled by a PC.
Extraction of DNA from spores
The extraction efficiency and purity of extracted DNA using the pretreatment kit are examined for 5x108 cfu of B. subtilis. The extraction efficiency of DNA was approximately 30%, and the ratio of absorbance (260 nm/280 nm) of extracted sample was around 2.0. These results indicated that the yield and purity of extracted DNA using the kit was sufficient for the molecular testing.
Sensitivity of system
After the extraction of DNA from B. anthracis using the pretreatment kit, we estimated the sensitivity of the Bio Bulwark system. Figure 4 shows a typical histogram of electrical signals for B. anthracis detection. Electrical signals from 3 genes related to B. anthracis are obviously higher than those from other genes. The dilution experiments showed that 10 pg gDNA and 1 pg pXO1 and 10 pg pXO2 were clearly identified by using Bio Bulwark. A genome size of B. anthracis is about 5.23 Mbp . These amounts of genes are equivalent to 10^3 cfu bacteria.
Specificity of system
B. anthracis has different pathogenicity depending on the presence of plasmid DNA pXO1 and/or pXO2. By using various B. anthracis strains, we have checked the specificity of our DNA chip cassette. For Pasteur I and Pasteur II (No.30) strains that have both pXO1 and pXO2, the currents for all three probes are positive. For Davis strain and Pasteur II (No.34) with only pXO2, the current for pXO1 is negative. For Pasteur II (No.36) without plasmid, the signal of gDNA is only positive.
We also tested cross-reactivity to 12 species (23 strains) closely related to B. anthracis, such as B. cereus and other Bacillus spp. The electrochemical currents for all 3 genes are negative to the related species. These results proved the high specificity for various anthrax strains.
Although we only describe the results of B. anthracis detection in this paper, the DNA chip cassette was not limited to B. anthracis. We have accomplished the detection of nineteen biological agents using this system.
We have developed a mobile system for biological agent detection, Bio BulwarkTM. Only by injecting a crude extract into a disposable cassette, the target genes of B. anthracis were easily detected within 70 min. This system makes it possible to detect biological agents on site with high sensitivity, specificity and rapidity. It is expected that this system would prevent or minimize damage in the event of a criminal attack involving a biological threat.
A part of this work is supported by a grant from the Core Research for Evolution Science and Technology (CREST), Project of Japan Science and Technology Agency (JST).
Make the best use of Scientific Research and information from our 700 + peer reviewed, Open Access Journals