Aurelia Magdalena Pisoschi*
University of Agronomic Sciences and Veterinary Medicine of Bucharest, Romania
Received Date: October 01, 2014; Accepted Date: October 04, 2014; Published Date: October 13, 2014
Citation: Pisoschi AM (2014) Electroanalytical Techniques for the Determination of Sulphite Preservative: An Editorial. Biochem Anal Biochem 3:e151. doi: 10.4172/2161-1009.1000e151
Copyright: © 2014 Pisoschi AM. This is an open-access article distributed under the terms of the Creative Commons Attribution License, which permits unrestricted use,distribution, and reproduction in any medium, provided the original author and source are credited.
Visit for more related articles at Biochemistry & Analytical Biochemistry
Sulphite, a commonly used preservative in foods and drinks is known for its reducing properties, as it can undergo oxidation to sulphate. Just as in the case of ascorbate, sulphite takes part in the anti-oxidant defence system that hinders the oxidative decay of food and drink components [1-8]. Sulfites, in their different forms, are present in: processed meats, wines, beer and cider, soft drinks and fruit juices, jams and jellies, dried and pickled fruits, shell fish, as well as processed food products for which protection against rancidity is required . It has been stipulated that the antibrowning activity of sulfite involves direct inhibition of polyphenol oxidase or reduction of quinone derivatives (formed by the oxidation of phenolics) to 1,2-dihydroxybenzene (catechol) derivatives .
Sulfur dioxide proved a promoter of the inflammation of the airways by neutrophil activation and is also involved in bronchoconstriction and worsening of the asthma pathology [9,10]. The adverse effects of sulphite may also include dermatitis, urticaria, flushing, hypotension, abdominal pain and diarrhea, or anaphylactic reactions , that is why its incorporation doses in the above-mentioned foodstuffs are rigorously reglemented by Codex Alimentarius .
Its reductive power and inherent electroactivity enables its determination through various electrochemical techniques, namely voltammetry, amperometry, potentiometry.
During the investigation of sulphite electrochemical behavior at the surface of a graphite electrode in alkaline solution, it has been assessed that sulphite oxidation is an irreversibile process .
2SO32- + O2 ? 2 SO42-
- sulphite anion is oxidized to sulphate in two successive steps, each implying one electron transfer, as follows :
SO32- ? SO3•- + e-
SO3•- + 2 OH- ? SO42- + H2O + e-
?sulphite anion is oxidized to sulphite radical in one single step characterized by one electron transfer, and then two sulphite radicals can interact giving dithionate S2O62-, which consecutively disproportionates, resulting in sulphate and sulphite [14-16]. If the process takes place in alkaline media, it can be described by the equations below :
2SO3•- ? S2O62-
S2O62- + 2OH- ? SO32- + SO42- + H2O
Voltammetry as electroanalytical technique involves recording of the current intensity, at an imposed potential variation. In the case of irreversible redox couples, the intensity of the cyclic voltammetric anodic oxidation peak can be directly correlated to the analyte concentration value.
Voltammograms corresponding to sulphite oxidation to sulphate in alkaline solution at a Pt stationary working electrode, show that the anodic oxidation peak becomes more pronounced and moves towards more positive potentials with increasing sweep rate that varied from 2 to 200 mV/s .
Sulphite electro-oxidation at a Pt disc rotating electrode has been reported to begin at around 380 mV and to attain a limiting plateau at approximately 1000 mV .
The much less figured cathodic peak (in comparison to the anodic one) noted at -230 mV, has been assigned to the reduction of Pt oxides formed at the electrode surface [14,17,18]. The lack of a reduction peak corresponding to sulphite in both cases (stationary and rotating disc Pt electrodes) could suggest that either sulphite oxidation is irreversibile, or that it may be followed by a reaction which swiftly eliminates the product from the electrode surface .
The figure below illustrates a cyclic voltammogram obtained for sulphite with a potentiostat-galvanostat, at a Pt working electrode (Aurelia Magdalena Pisoschi, unpublished results).
A novel cyclic voltammetric sensor for the simultaneous determination of SO32– and NO2– was proposed, relying on a glassy carbon electrode modified with graphene-chitosan and gold nanoparticles. The linear analytical range for sulphite corresponded to 5 µM-0.41mM, at 1µM detection limit. Moreover, the relative standard deviation value for sulphite was 1.1%, the developed modified sensor being characterized by sensitivity, stability and reproducibility .
Amperometry is based on the recording of the current intensity at a fixed applied potential, in the absence or presence of a biorecognition element (incorporated in amperometric biosensors). The amperometric detection of sulphite in the presence of iodate and periodate, at a Pt electrode modified with multi-walled carbon nanotubes and Prussian blue nanoparticles, was possible at 600 mV potential value for sulphite, and 320 mV for iodate and periodate. The detection limits and catalytic rate constants were 4.70 µM (5.24×107 cm3 mol-1 s-1), 8.30 µM (1.37×106 cm3 mol-1 s-1) and 8.06 µM (0.34×106 cm3 mol-1 s-1), for sulphite, iodate and periodate, respectively .
A sulfite biosensor was developed by immobilizing the enzyme sulfite oxidase at glassy carbon electrode coated with a mercury film. The electrochemical reduction of molecular oxygen at the surface of the working electrode is the reaction generating the analytical signal, at ?240 mV working potential vs. Ag/AgCl reference . Essentially, the rate of oxygen consumption was proportional to the sulfite amount, and the peak current intensity decreased owing to molecular oxygen depletion in the sulphite oxidase catalyzed reaction:
SO32- + O2 + H2O ? SO42- + H2O2
The analytical linear response range was obtained between 2×10-4 and 2.8×10-3 M sulfite .
Potentiometry relies on recording the potential difference that occurs as a result of an ionic species concentration (activity) change. The potential variation can be the outcome of a shift in the ratio between the oxidized and the reduced form of an indicating species that eventually transduces itself as a potential difference. For the assessment of sulphite (a reductant), the higher the concentration of the analysed species, the greater the concentration of the reduced form of the indicator, and therefore, the more enhanced the potential change. It has been determined that the use of naphtoquinone as indicator, results in good sensitivity towards sulphite .
It is important to stress out the role of the investigation and optimization of performant analytical methods, regarding food additives for which the content has to be rigorously controlled. Electrode modification may lead to enhanced analytical performances in terms of sensitivity and reproducibility; nevertheless, viable results can also be obtained with bare electrodes.
Make the best use of Scientific Research and information from our 700 + peer reviewed, Open Access Journals