Author(s): van der Pol E, Coumans FA, Grootemaat AE, Gardiner C, Sargent IL,
Abstract Share this page
Abstract BACKGROUND: Enumeration of extracellular vesicles has clinical potential as a biomarker for disease. In biological samples, the smallest and largest vesicles typically differ 25-fold in size, 300,000-fold in concentration, 20,000-fold in volume, and 10,000,000-fold in scattered light. Because of this heterogeneity, the currently employed techniques detect concentrations ranging from 10(4) to 10(12) vesicles mL(-1) . OBJECTIVES: To investigate whether the large variation in the detected concentration of vesicles is caused by the minimum detectable vesicle size of five widely used techniques. METHODS: The size and concentration of vesicles and reference beads were measured with transmission electron microscopy (TEM), a conventional flow cytometer, a flow cytometer dedicated to detecting submicrometer particles, nanoparticle tracking analysis (NTA), and resistive pulse sensing (RPS). RESULTS: Each technique gave a different size distribution and a different concentration for the same vesicle sample. CONCLUSION: Differences between the detected vesicle concentrations are primarily caused by differences between the minimum detectable vesicle sizes. The minimum detectable vesicle sizes were 70-90 nm for NTA, 70-100 nm for RPS, 150-190 nm for dedicated flow cytometry, and 270-600 nm for conventional flow cytometry. TEM could detect the smallest vesicles present, albeit after adhesion on a surface. Dedicated flow cytometry was most accurate in determining the size of reference beads, but is expected to be less accurate on vesicles, owing to heterogeneity of the refractive index of vesicles. Nevertheless, dedicated flow cytometry is relatively fast and allows multiplex fluorescence detection, making it most applicable to clinical research. © 2014 International Society on Thrombosis and Haemostasis.
This article was published in J Thromb Haemost
and referenced in Journal of Biotechnology & Biomaterials